We are probably going to start using this too. I believe Cleen is the new name for Fleet enema.
Lizzie Lindley is probably the person to start with since she was working on draft guidelines for this sort of thing.
thanks for the replies gents, we've been using it for a while now with no side affects on patients but a few of our AK95's are giving us problems with conductivity which dissappears when we change the concentrate for fresh non additive cans. For your info Chris we add it to concentrate to boost the patients Phosphate levels
This message may contain confidential information. If you are not the intended recipient please inform the
sender that you have received the message in error before deleting it.
Please do not disclose, copy or distribute information in this e-mail or take any action in relation to its contents. To do so is strictly prohibited and may be unlawful. Thank you for your co-operation.
NHSmail is the secure email and directory service available for all NHS staff in England and Scotland. NHSmail is approved for exchanging patient data and other sensitive information with NHSmail and other accredited email services.
For more information and to find out how you can switch, https://portal.nhs.net/help/joiningnhsmail
I think all our data was collected during HDF treatments Chris. I know the paper that was done mentions HD but I don't think it was specifically meaning the it was only done on standard HD mode. As HDF replacement is just filtered dialysate anyway the phosphate rich fluid is being infused directly into the patient bloodstream if you were perhaps worried about it being dialysed out or something?
I've just had a reply from MTN and they had already been asked by another customer and unfortunately have said no because of the additional costs of all the solution testing.
Perhaps if the numbers grow considerably they might reconsider.