I have a problem with a GAMBRO machine. In preparation mode, the level detector in the degassing chamber is too low. Same thing for the level in dialysate chamber. I've replaced the level detector. The degassing pump is running well.
Any suggestions can help me.
Is it drawing fluid round from the heating vessel?
Pop the tube off from the top of the filter and see if it has suction there. If yes try the T-piece on the heating vessel. If not repeat until you get to the degas pump. T-piece on top of Pump is also common.
Guys and St Thomas NHS Foundation Trust